Innovation of Dental X-Ray Holder Using Silicone Rubber Coating in Posterior Dental Periapical Intraoral Examination
The major drawback of the parallel periapical examination technique is that the holder used can damage the oral tissues and cause discomfort to the patient. The objective of this study is to determine the work efficiency and radiographic quality of the innovative dental x-ray holder which has been made by adding synthetic rubber or silicone to the part of the holder that is in direct contact with the patient. This research is an experimental with a post-test only design. The analysis was performed based on filling out the questionnaire on a Likert scale ranging from 1 to 4. With the criteria 1. Disagree, 2. Sometimes, 3. Agree and 4. Strongly Agree. The test was administered by comparing the holder made with the commonly used Aphrodite holder as a control group. There were 16 repetitions of exposure to the cadaveric skull in obtaining research data for each treatment group. The results of statistical work efficiency testing on the control group resulted in a value of B = 0.125 with a significance of 0.071 and an effect of 10.5%. Meanwhile, for testing the quality of radiographic image, the value of B = 0.125 with a significance of 0.014 and an effect of 18.5% was obtained. The innovative dental x-ray holder using a silicone rubber layer is efficient and the resulting radiographic image quality is good when used in the intraoral examination.
Aps, J. K. M., Lim, L. Z., Tong, H. J., Kalia, B., & Chou, A. M. (2020). Diagnostic efficacy of and indications for intraoral radiographs in pediatric dentistry: a systematic review. European Archives of Paediatric Dentistry, 21, 429-462. doi: https://doi.org/10.1007/s40368-020-00532-y
Arx, T. V., & Lozanoff, S. (2016). Clinical oral anatomy: A comprehensive review for dental practitioners and researchers. Clinical Oral Anatomy: A Comprehensive Review for Dental Practitioners and Researchers. Switzerland: Springer International Publishing. doi: https://doi.org/10.1007/978-3-319-41993-0
Cowan, I. A., MacDonald, S. L., & Floyd, R. A. (2013). Measuring and managing radiologist workload: Measuring radiologist reporting times using data from a Radiology Information System. Journal of Medical Imaging and Radiation Oncology, 57(5), 558–566. doi: https://doi.org/10.1111/1754-9485.12092
Dardengo, C. D. S., Fernandes, L. Q. P, & Júnior, J. C. (2016). Frequency of orthodontic extraction. Dental press journal of orthodontics, 21(1), 54-59. doi:https://doi.org/10.1590/2177-6709.21.1.054-059.oar
El-Angbawi, A. M. F., McIntyre, G. T., Bearn, D. R., & Thomson, D. J. (2012). Film and digital periapical radiographs for the measurement of apical root shortening. Journal of Clinical and Experimental Dentistry, 4(5), e281. doi: https://doi.org/10.4317/jced.50872
Gupta, A., Devi, P., Srivastava, R., & Jyoti, B. (2014). Intra oral periapical radiography - basics yet intrigue: A review. Bangladesh Journal of Dental Research & Education, 4(2), 83–87. doi:https://doi.org/10.3329/bjdre.v4i2.20255
Iannucci, J., & Howerton, L. J. (2016). Dental Radiography-E-Book: A Workbook and Laboratory Manual. St. Louis, Missouri: Elsevier.
Ilhan, B., Bayrakdar, İ. S., & Orhan, K. (2020). Dental radiographic procedures during COVID-19 outbreak and normalization period: recommendations on infection control. Oral Radiology, 36(4), 395–399. doi: https://doi.org/10.1007/s11282-020-00460-z
Khorasani, M. M. Y., & Ebrahimnejad, H. (2017). Comparison of the accuracy of conventional and digital radiography in root canal working length determination: An invitro study. Journal of dental research, dental clinics, dental prospects, 11(3), 161-165. doi: https://doi.org/10.15171/joddd.2017.029
Manja, C. D., & Fransiari, M. E. (2018). A comparative assessment of alveolar bone loss using bitewing, periapical, and panoramic radiography. Bali Medical Journal, 7(3). doi: https://doi.org/10.15562/bmj.v7i3.1191
Mojsiewicz-Pieńkowska, K., Jamrógiewicz, M., Szymkowska, K., & Krenczkowska, D. (2016). Direct human contact with siloxanes (silicones)–safety or risk part 1. Characteristics of siloxanes (silicones). Frontiers in pharmacology, 7, 132. doi: https://doi.org/10.3389/fphar.2016.00132
Monika, A. P. W., Astuti, E. R., & Mulyani, S. W. M. (2020). The Quality of The Lollipops Use in The Making of The Anterior Upper Teeth Periapical Radiography of in Paediatric Patients. Eurasian Journal of Biosciences, 14, 4049–4053. Retrieved from http://ejobios.org/download/the-quality-of-the-lollipops-use-in-the-making-of-the-anterior-upper-teeth-periapical-radiography-of-8047.pdf
Pando, J.A.G., Sainz, Z. de la C. T., Reyes, J. C., Concepcion, J. A. C., Santos, I. F. (2019). Effectiveness of Periapical Radiographic Methods by Parallelism and Bisection. Rev Ciencias Médicas, 23(5), 654–663.Retrieved from https://www.medigraphic.com/pdfs/pinar/rcm-2019/rcm195h.pdf
Reynolds, T. (2016). Basic Guide to Dental Radiography. Wiley Blackwell.
Whaites, E., & Drage, N. (2013). Essentials of dental radiography and radiology (5th ed.). Churchill Livingstone.
Whitley, A. S., Jefferson, G., Holmes, K., Sloane, C., Anderson, C., & Hoadley, G. (2015). Clark’s Positioning in Radiography (A. D. Moore (ed.); 13th ed.). CRC Press.
Copyright (c) 2021 JURNAL INFO KESEHATAN
This work is licensed under a Creative Commons Attribution-NonCommercial-ShareAlike 4.0 International License.
Ownership of copyright
The copyright in this website and the material on this website (including without limitation the text, computer code, artwork, photographs, images, music, audio material, video material and audio-visual material on this website) is owned by JURNAL INFO KESEHATAN and its licensors.
JURNAL INFO KESEHATAN grants to you a worldwide non-exclusive royalty-free revocable license to:
- view this website and the material on this website on a computer or mobile device via a web browser;
- copy and store this website and the material on this website in your web browser cache memory; and
- print pages from this website for your use.
- All articles published by JURNAL INFO KESEHATAN are licensed under the Creative Commons Attribution 4.0 International License. This permits anyone to copy, redistribute, remix, transmit and adapt the work provided the original work and source is appropriately cited.
JURNAL INFO KESEHATAN does not grant you any other rights in relation to this website or the material on this website. In other words, all other rights are reserved.
For the avoidance of doubt, you must not adapt, edit, change, transform, publish, republish, distribute, redistribute, broadcast, rebroadcast or show or play in public this website or the material on this website (in any form or media) without appropriately and conspicuously citing the original work and source or JURNAL INFO KESEHATAN prior written permission.
You may request permission to use the copyright materials on this website by writing to email@example.com.
Enforcement of copyright
JURNAL INFO KESEHATAN takes the protection of its copyright very seriously.
If JURNAL INFO KESEHATAN discovers that you have used its copyright materials in contravention of the license above, JURNAL INFO KESEHATAN may bring legal proceedings against you seeking monetary damages and an injunction to stop you using those materials. You could also be ordered to pay legal costs.
If you become aware of any use of JURNAL INFO KESEHATAN copyright materials that contravenes or may contravene the license above, please report this by email to firstname.lastname@example.org
If you become aware of any material on the website that you believe infringes your or any other person's copyright, please report this by email to email@example.com.